METER TEROS 10 Soil Moisture Sensor User Guide
TEROS 10 QUICK START
Inspect and verify TEROS 10 components. The TEROS verification clip gives the best assessment of proper sensor function and accuracy. The TEROS 10 should read 0.35 to 0.42 m3/m3 on the verification clip. If a verification clip is not available, test basic sensor functionality in air and water. The TEROS 10 will read ~0.64 m3/m3 in water and a slightly negative value in air.NOTE: The sensors are optimzed to read in soils, therefore the sensor will not read 100% in pure liquid water. Values above use the mineral soil calibration.
Proper installation of the sensors is critical for proper operation. Refer to the TEROS 10 User Manual for details.For easy installation, use the borehole installation tool. The installation tool (shown below) is available for rent from METER Group. Contact Customer Support for more information.
What is soil moisture?
Soil moisture is a key variable in controlling the exchange of water and heat energy between the land surface and the atmosphere through evaporation and plant transpiration.
For best results, use the latest versions of METER software and firmware for the computer or mobile device, ZENTRA products, and sensors. Please use the software Help menu to find updates.Consult the sensor user manual for more troubleshooting tips..
- Insert Sensor Auger or trench a hole to the desired sensor depth. Insert the sensor into the undisturbed soil.When using the borehole installation tool, load the TEROS 10 as shown. Lower the tool into the hole or trench with the back of the tool supported by the far wall. Pull on the lever to activate the jack and insert the sensor into hole wall.
- Check Sensor Operation Plug the sensor into the data logger and use the SCAN function in the software to do a quick check of sensor operation before backfilling.
- Repack Soil and Protect Cables Secure and protect cables with PVC casing or flexible conduit and backfill the trench or hole.
- Plug Sensor In and Configure Logger Plug the sensor into the data logger. Use data logger software to apply appropriate settings to the sensors plugged into each data logger port.The TEROS 10 is an analog sensor and will not be automatically recognized on METER data loggers. Select the right sensor for each port.
Have a question or problem? Our support team can help.We manufacture, test, calibrate, and repair every instrument in house. Our scientists and technicians use the instruments every day in our product testing lab. No matter what your question is, we have someone who can help you answer it
Email: Phone: +1.509.332.5600
Email: Phone: +49 89 12 66 52 0
Read More About This Manual & Download PDF: